Rapid metabolic phenotyping in SARS-CoV-2 patients using the impact II QTOF MS system
Nicola Gray, Australian National Phenome Centre and Center for Computational and Systems Medicine, Murdoch University
Nathan G. Lawler, Australian National Phenome Centre and Center for Computational and Systems Medicine, Murdoch University
Rongchang Yang, Australian National Phenome Centre, Murdoch University
Aude-Claire Morillon, Australian National Phenome Centre, Murdoch University
Melvin C.L. Gay, Bruker Pty Ltd, Australia
Sze-How Bong, Australian National Phenome Centre, Murdoch University
Elaine Holmes, Australian National Phenome Centre and Center for Computational and Systems Medicine, Murdoch University; Section for Nutrition Research, Imperial College London
Luke Whiley, Australian National Phenome Centre and Center for Computational and Systems Medicine, Murdoch University; The Perron Institute for Neurological and Translational Science
Jeremy K. Nicholson, Australian National Phenome Centre and Center for Computational and Systems Medicine, Murdoch University; Institute of Global Health Innovation, Imperial College London
Aug 23, 2021
The impact II QTOF system meets the critical elements of sensitivity, selectivity, accuracy, and precision for targeted metabolite analysis. The continuous collection of both MS and MS/MS data provides an exponentially deeper and broader pool of data valuable for subsequent retrospective discovery or targeted analyses. This analytical flexibility is particularly valuable in metabolic phenotyping of biological samples when molecular mechanisms are poorly understood and knowledge is emerging at a rapid pace, as with the SARS-CoV-2 global pandemic.
The impact II QTOF MS system has been established to have high utility in many critical analyses, including toxicology, forensic medicine, antidoping, and pesticide screening [1-3]. This application flexibility is driven in part by the system's robustness and ability to simultaneously record full, high accurate mass, MS scan data together with accurate mass MS/MS fragmentation information for confident identification of both known and novel targets. Further, robust hardware and software components support high sample throughput and subsequent rapid data processing requirements.
Within the “’omics” scope, interest in metabolic phenotyping has dramatically increased in recent years. Individual metabolic phenotypes, determined by a complex history of gene-environment interactions, may be linked to disease risks , and population-based profiling has been widely applied to expand the biochemical knowledge of a range of diseases as well as in effort to understand patterns of disease severity. However, a targeted approach of quantitating particular metabolite panels linked to a given disease state or a metabolic pathway of interest can provide detailed insights to support biomedical research or to specifically characterize an individual’s current state of health. This workflow often employs LC-TQ MS systems due to their low detection limits and wide dynamic range in targeted analyses.
In this study, the quantitative performance of the impact II UHR-QTOF (Ultra-High Resolution Quadrupole-Time-Of-Flight) for metabolite analyses was compared to that of a current TQ MS system in the quantitative detection of 34 biogenic amines reported to be altered in COVID-19 positive individuals . Three human biofluids (plasma, serum, and urine) were profiled on multiple instruments (two impact II QTOF systems, three TQ systems) within a certified laboratory (The Australian National Phenome Centre, ANPC). A certified external reference sample (NIST SRM 1950 plasma) was also profiled on all systems.
The elements of analytical method sensitivity, detection accuracy, and reproducibility, as examined, are critical for confidence in any screening workflow, including metabolic phenotyping. The extension of analytical scope within the data collected is also of high value in order to fully explore any unique biochemical signatures, particularly in the case of new or emerging clinical challenges. In this study, the accurate mass MS and MS/MS data sets provided by the impact II systems could be retrospectively examined to identify an additional bioamine of clinical significance.
Briefly, 10 μL of biofluids (plasma, serum, and urine) were centrifuged and transferred onto a 96-well plate. For the analysis of the cohorts of SARS-CoV-2 (positive and negative) plasma samples, a pooled sample was also prepared. Each plate included calibrators (1 to 400 μmol/L) and quality controls (QC; 300, 75, 15, and 3 μmol/L) prepared in DI water. An internal standard mixture was added to each well, followed by protein precipitation with methanol. An aliquot of the supernatant was transferred into a new 96-well plate and derivatized using 6-aminoquinolyl-N-hydroxysuccinimidyl carbamate to transform primary and secondary amines into highly stable derivatives. The subsequent derivatized samples were diluted 1:4 and 1:49 with water for impact II QTOF and TQ analysis, respectively. Refer to reference  for detailed methodology.
Instrument parameters are listed in Table 1.
QTOF MS and MS/MS data from the impact II data were processed using TASQ® 2.2. The molecular formula or exact mass of the derivatized amino acid was used to extract the precursor ions with a mass error of < 3 mDa. Calibration curves were linearly fitted with a weighting factor of 1/x. A representative QTOF Extracted Ion Chromatogram is shown in Figure 1. MRM data collected on the TQ systems was processed using the manufacturer’s software, and calibration curves were linearly fitted with a weighting factor of 1/x2.
A multidimensional evaluation of the analytical performance of the impact II QTOF systems was conducted using the QC samples and compared directly with the performance of the TQ systems. Metabolite target selectivity, calibration linearity, intra- and inter-day accuracy, precision, and matrix effects were each examined, based on a previously reported protocol adapted from FDA and EMA bioanalytical guidelines [7,8]. The quantitation accuracy of the method was investigated with the analysis of reference plasma (NIST SRM 1950) for comparison to the certified reference values for 12 available amino acids.
Authentic biofluid analysis
For a more comprehensive comparison between the two analytical platforms, 34 target biogenic amines (Table 2) were sought in 12 individual healthy human plasma, serum, and urine samples, with data collection as previously defined for each platform. Further statistical analyses were applied identically to all data sets using ANPC’s in-house pipelines.
For a broader, more systemic comparative view, targeted quantitative metabolite analyses were made on both the QTOF and TQ platforms on plasma samples from a cohort of individuals infected with SARSCoV-2 (n=7) and a cohort of healthy controls (n=8). Unlike previous comparisons, the levels of the 34 target biogenic amines as detected on both platforms were compared together within a multivariate PCA. Reference plasma (NIST SRM 1950) was also analyzed as a control.
Retrospective data mining
Retrospective analysis of the QTOF MS data for additional analytes of interest (not originally considered in the targeted comparative profiling) was initiated by the analysis of the pooled plasma sample from the cohort study using the auto MS/MS method (Table 1) and processed on MetaboScape®2021. A table of mass features, or ‘buckets,’ was generated, grouping together all related adducts (e.g., [M+H]+, [M+Na]+, etc.) for all detected precursor compounds. To allow easier viewing of derivatived compounds, the dataset was filtered within MetaboScape to show/display only buckets containing MS/MS fragments of the AQC adducts (C8H10N2O2; 171.055 m/z), which is a common fragment signature from the cleavage at the ureide bond formed upon derivatization. Subsequently, all buckets from the previously targeted biogenic amines were excluded. Compounds that were not annotated were further investigated using accurate mass MS and MS/MS data for tentative identification using SmartFormula. The identity of one compound (tentatively identified) to be of clinical interest was confirmed by comparison with a commercial standard.
Results and Discussion
Comparative analytical performance for method validation
TQ MS systems are often the instrument of choice for routine targeted and quantitative workflows as they are generally considered to be more sensitive and robust than QTOF MS platforms. In this study, the absolute lower limits of quantitation (LLOQ) for all targeted amino acids and bioamines by the QTOFs (400 nmol on-column) were higher than in the TQ (40 nmol on-column). However, the multiple reaction monitoring (MRM) of the TQ systems was challenged by an unknown co-eluting matrix interference in two of three monitored transitions for the arginine target in many of the urine samples (Figure 2A). This interference was not detected in the QTOF system. The high mass accuracy of the impact II permits the application of a narrow mass extraction window (± 0.005 Da), as shown in the extracted ion chromatogram (EIC) (Figure 2B). All target compounds could be quickly and cleanly isolated from the complete data pool, regardless of sample type.
The impact II QTOF MS demonstrated excellent quantitative performance with linearity over the concentration range (1 – 400 μmol/L) with correlation coefficients (r2) higher than 0.99 and residuals lower than 15% for all metabolite targets, similar to those observed in TQ.
Accuracy and Precision
The intra- and inter-run accuracy, expressed as mean percentage bias (%), and precision, expressed as coefficient of variance (% CV), were evaluated based on the recommendations of the European Bioanalysis Forum  for metabolites in plasma samples. The mean bias of the 34 amino acids was within ± 20% in analyses on both system types, as were the % CV values for three replicates of the four QC levels, as demonstrated by the radar plots in Figure 3A.
Through the parallel analyses of the NIST SRM 1950 standard on both platforms, the quantitation accuracy of 12 amino acids was found to be in good agreement with the certified reference values. The accuracy of the concentrations observed by the instruments was within 12.9% bias for all 12 amino acids. Values for alanine, glycine, and histidine are shown in Figure 3B.
Sample matrices are known to cause ion suppression in many MS analyses. To evaluate this effect, 24 stable isotope labeled (SIL) internal standards were spiked into individual matrices (serum, plasma, and urine) or reference solution. The matrix effect (% ME), expressed as a percentage, illustrates the degree of ion suppression/enhancement as shown in Figure 4. Of the 24 SIL investigated, only cystine in urine was found to be adversely affected by ion suppression (>25%). Further investigation using the high-resolution accurate mass full scan data predicted a molecule with a chemical formula of C10H9N3O (m/z 188.0824 [M+H+], which was identified as the 6-aminoquinoline derivative of ammonia. This finding illustrates a key advantage of full scan accurate mass data acquisition of QTOF instruments – the ability to detect and identify co-eluting species.
Quantitative robustness and parity for authentic biofluids
The robustness of the methodologies was further investigated with three common types of biofluids: serum, plasma, and urine (n=12 each). A high level of agreement was observed between the calculated amino acid concentrations in serum, plasma, and urine as measured across the various platforms, with a correlation coefficient of >0.85 (Figure 5). Furthermore, Bland-Altman analysis demonstrated that all measurements were also within the limits of agreement for all platforms.
As mentioned, the 34 metabolic targets evaluated within the course of this work had been previously shown to be modified in SARS-CoV-2 positive individuals . In some cases, monitoring individual metabolites may be of particular interest, potentially serving as analytical guideposts for specific metabolic pathways or organ function. Concentration differences in several metabolites demonstrated to be altered by SARS-CoV-2 [5,9,10,11], such as tryptophan and taurine, were also observed in plasma from the cohort of individuals who had tested positive for SARS-CoV-2 and the cohort of healthy controls (Figure 6). In this study, the small sample cohort size precludes rigorous evaluation of statistical significance; however, the trends of these differentiating metabolites are evident and independent of the analytical platform.
Multivariate analysis of the 34 targeted bioamines and amino acids in the SARS-CoV-2 positive and negative cohorts (Figure 7) further illustrates the robustness of the QTOF MS system. Individual patient samples cluster similarly on both platforms; thus, the separation of diseased and healthy cohorts was equitable between the QTOF (data shown as triangles) and TQ (shown as circles) MS systems. This analysis also included replicates of the certified reference plasma, which clustered very tightly together (Figure 7, red triangles and circles). This discriminatory parity between the two platforms is an important factor to consider, as metabolites may be able to serve as diagnostic biomarkers or help to guide care based on identified systemic effects.
Discovery metabolomics (Retrospective data mining)
As the data collected on the impact II QTOF system 1) was untargeted, and 2) included both high mass accurate MS and MS/MS data, analytes that were not originally considered in targeted assay development could be retrospectively sought. In a subsequent analysis of the pooled plasma samples from the SARS-CoV-2 cohorts, an additional 2700 features were detected using MetaboScape 2021. The feature list was further reduced using the unique MS/MS fragment of the AQC adducts and annotated with derivatized amino acid Analyte List (Figure 8A). With the simultaneous acquisition of high accuracy mass measurement of the MS and MS/MS fragmentation data, among these “un-annotated” features was kynurenine (C20H18N4O4, theoretical mass m/z 379.1401 ([M+H+]), which has been shown to be altered in SARS-CoV-2 individuals . Its chemical formula was accurately predicted using Smart Formula and confirmed with a commercial kynurenine standard (Figure 8B and 8C).
- The precision and accuracy of the impact II QTOF system in the analysis of reference samples were sufficient to meet current metabolite profiling recommendations and were comparable to a contemporary TQ platform. Performance was robust and reliable, suitable to support the sample throughput (around 190 samples/day) necessary for the study of larger sample cohorts.
- High accurate mass MS detection on the QTOF system enabled narrow mass window filtering to eliminate co-eluting matrix interference, which prohibited clear detection of arginine in urine samples within the TQ workflow. Further, the full scan data collection of the QTOF permitted identification of a co-eluting ammonia derivative which had suppressed the detection of cystine in urine in comparative matrix effect evaluations.
- The simultaneous collection of full scan (complete m/z range) accurate mass MS and MS/MS data utilized within the QTOF workflow offers unique flexibility and analytical power. MS and MS/MS data for kynurenine, previously reported to be altered by SARS-CoV-2, were extracted from the complete data pool.
- Multivariate profiling of target metabolites previously found to be linked to SARS-CoV-2 was comparable in the analysis of true human samples (SARS-CoV-2 positive versus healthy controls), reinforcing the utility of the QTOF platform within the metabolic phenotyping community.
 Bruker Technical Note TN-47: Proven robustness for large-scale metabolomics studies using the ‘Phenomics Workhorse’
 Bruker Application Note LCMS 168: Urine Screening using online SPE-LC-MS.
 Bruker Application Note LCMS 140: Quantitation of Drugs of Abuse in Oral Fluid.
 Holmes E, Wilson ID, Nicholson JK (2008). Metabolic Phenotyping in Health and Disease. Cell. 134(5):714-7. doi: 10.1016/j.cell.2008.08.026.
 Kimhofer T, Lodge S, Whiley L, et al. (2020). Integrative Modeling of Quantitative Plasma Lipoprotein, Metabolic, and Amino Acid Data Reveals a Multiorgan Pathological Signature of SARS-CoV-2 Infection. J Proteome Res. 19(11):4442-4454. doi:10.1021/acs.jproteome.0c00519
 Gray N, Lawler NG, Yang R, et al. (2021) A simultaneous exploratory and quantitative amino acid and biogenic amine metabolic profiling platform for rapid disease phenotyping via UPLC-QToF-MS. Talanta. 223(Pt 2):121872. doi:10.1016/j.talanta.2020.121872
 Gray N, Zia R, King A, et al. (2017). High-Speed Quantitative UPLC-MS Analysis of Multiple Amines in Human Plasma and Serum via Precolumn Derivatization with 6-Aminoquinolyl-N-hydroxysuccinimidyl Carbamate: Application to Acetaminophen-Induced Liver Failure. Anal Chem. 89(4):2478-2487. doi:10.1021/acs.analchem.6b04623
 Gray N, Plumb RS, Wilson ID, Nicholson JK (2019). A validated UPLC-MS/MS assay for the quantification of amino acids and biogenic amines in rat urine. J Chromatogr B Analyt Technol Biomed Life Sci. 1106-1107:50-57. doi:10.1016/j.jchromb.2018.12.028
 Timmerman P, Blech S, White S, et al. (2016). Best practices for metabolite quantification in drug development: updated recommendation from the European Bioanalysis Forum. Bioanalysis. 8(12):1297-1305. doi:10.4155/bio-2016-0103
 Barberis E, Timo S, Amede E, et al. (2020). Large-Scale Plasma Analysis Revealed New Mechanisms and Molecules Associated with the Host Response to SARS-CoV-2. Int J Mol Sci. 21(22):8623. doi:10.3390/ijms21228623.
 Thomas T, Stefanoni D, Reisz JA, et al. (2020). COVID-19 infection alters kynurenine and fatty acid metabolism, correlating with IL-6 levels and renal status. JCI Insight. 5(14):e140327. Published 2020 Jul 23. doi:10.1172/jci.insight.140327